MEASUREMENT MICROSCOPE GENERAL SERVICE
-- service procedure (optek/toolserve-001 Rev. 004 6/18/99) --
Perform an initial evaluation including photo
and video system, DRO, encoders stage, hardware, seals, adjustment etc….A
written report is available upon request.
2.0 Clean exterior surfaces or microscope and related components as necessary.
3.0 Clean optical elements including eyepieces, beam splitter mirror, objectives and final magnification lenses.
4.0 Check stage operation. Adjust as necessary.
5.0 Clean and lubricate stage, focus mechanism and all sliding wear surfaces and mechanisms as necessary.
6.0 Check operation of the lens turret. Make adjustments as necessary.
7.0 Clean illumination system.
8.0 Adjust illumination system as necessary for best image sharpness and quality.
9.0 Clean, center, focus and adjust photo and /or video system as required.
10.0 Check hardware and fasteners for looseness. Tighten as necessary.
11.0 Perform minor mechanical adjustments to optimize performance of microscope and accessories as necessary.
12.0 Report any out of spec. conditions of equipment to customer. Note any actions and or parts required to restore equipment to factory spec’s. Quote repair parts and labor cost upon request.
390 Bethany Drive, Scotts Valley, CA. 95066 USA
Phone: (831)438-4999 Fax: (831)438-1436
Copyright © 2001 OPTEKUSA.COM