STEREO AND STEREOZOOM MICROSCOPE GENERAL SERVICE
-- service procedure (optek/zoomserve-003 Rev. 003 8/15/97) --
Perform an initial evaluation. A written report is available upon
Clean exterior surfaces or microscope and
related components as necessary.
Clean optical elements including eyepieces,
objectives and final magnification lenses.
Check and adjust parfocality as necessary.
Reset diopter rings to zero.
Check for concentric alignment of fields of
view. Note: Minor adjustments of problems are included. Major adjustments
may require repair at additional cost. This is usually only necessary when the
scope is not usable or itís performance is being complained about.
Clean and lubricate stage, focus mechanism and
all sliding wear surfaces and mechanisms as necessary.
Clean and adjust illumination system as
Clean, center, focus and adjust photo and /or
video system as required.
Check hardware and fasteners for looseness.
Tighten as necessary.
Perform minor mechanical adjustments to
optimize performance of microscope and accessories as necessary.
11.0 Report any out of spec. conditions of equipment to customer. Note any actions and or parts required to restore equipment to factory specís. Quote repair parts and labor cost upon request.
390 Bethany Drive, Scotts Valley, CA. 95066 USA
Phone: (831)438-4999 Fax: (831)438-1436
Copyright © 2001 OPTEKUSA.COM